Validating auto blood pressure monitors Massivecams
In fact, there is a movement to create a universal standard for clinical validation. Some of the basic requirements are that the validation is performed by an independent party, using one of the established protocols (ex. Validation of the A&D BP UA-651 Device with a Wide-Range Cuff for Home Blood Pressure Measurement According to the European Society of Hypertension International Protocol (Revision 2010). To achieve a grade, all three percentages must be equal to or greater than the tabulated values. A/A): one for the systolic reading and the other for the diastolic reading. The ESH International Protocol and ISO Standard use a Pass/Fail grading system. Validation of the A&D UM-211 Device for Office Blood Pressure Measurement According to the European Society of Hypertension International Protocol Revision 2010. Due to necessary scheduled maintenance, the JMIR Publications website will be unavailable from Monday, March 11, 2019 at PM to PM EST.
The clinical validation tests verify the accuracy of the device according to certain criteria, like whether they have high, low, or normal blood pressure, and pre-existing diseases.